Company Statement" Zhengzhou KJ Technology Co., Ltd. " belongs to the international business department of KJ Group. Starting today, the company will continue the international business contract signed or currently being performed by " Zhengzhou Kejia Furnace Co., Ltd. " See details >>
PECVD system, by ionizing atom-containing gas with microwave or radio frequency, creates active plasma locally. Compared with traditional CVD system, it will react easily to deposit and form the expected thin film in low temperature. It widely used to pro
|Vertical PECVD System
|Alumina ceramic fiber
|Outside Shell Temperature
|20℃/min (Suggestion 10℃/min)
|Heating Zone Length
|440mm (Single Zone)
|0Cr27Al7Mo2 resistance wire
|Temperature Control Accuracy
Φ50, 60, 80, 100, 150mm available.
|PID automatic control with 30 programmable segments for precise control
|Flanges made of 304SS and silicon rubber rings
Gas valve on left flange
Vacuum angle valve with KF25 connector on the right flange
Note: Standard flange, KF flange and hinge flange are available.
|Single phase, 220V, 50Hz.
|Vacuum pump unit
(With movable box)
|10pa with a mechanical vacuum pump
0.001pa with a mechanical vacuum pump and a diffusion vacuum pump.
|Inficon Digital Pressure Gauge with high accuracy
Measurement Range 1×105--1×10-5Pa
|Plasma RF Power Supply
|Output Power Range
|13.56 MHz ±0.005% stability
|RF Output Port
|50 Ω, N-type, female
|208-240VAC, Single Phase, 50/60Hz
|Gas-Mixing Unit with mass flow meters
(Float flow meters are available.)
|Three precise mass flow meters (0.02% accuracy) are installed to control gas flow rate with touchscreen control for easy operation.
One gas mixing tank is installed on the bottom case.
MFC 1: Gas flow range from 0~100 SCCM
MFC 2: Control range from 0~200 SCCM
MFC 3: Control range from 0~500 SCCM
Gas inlet fitting: four 1/4NPS.
Gas outlet fitting: four 1/4NPS.
Power: 185-220VAC 50/60Hz
4 stainless steel needle valves are installed on the left side to manually control the inlet and outlet of gas.
|Be used to cool the tube and vacuum pump.
|Pressure higher than 1.02Mpa will break the tube.
Since the pressure in the gas cylinder is high, a converter is necessary. The measurement range of our converter is 0.01mpa-0.1mpa, will ensure a safe gas supply.
No vacuum condition is allowed when the temperature is higher than 1000℃.
Please keep the gas flow lower than 200sccm when the temperature is high to lower the impact of cold air to the hot tube.
If you have to keep the valves on the two ends of the tube closed at the same time, please watch the pressure gauge and open the valves once the pressure is higher than 1.02Mpa to avoid any tube breakage.
|One year with lifetime technical support.
Please leave a message here. We will reply you in 24 hours.
Wechat/Whatsapp: +86 18037178440
Company Tell: +(86)18037178440